Physical testing of TKR systems to assess stability is an important aspect in screening candidate TKR designs which can be expensive and time consuming. Costs can be reduced by utilizing 3D printed plastic components. The objective is to compare the kinematics and intrinsic constraint of metal-on-plastic (M-P) and plastic-on-plastic (P-P) implants under physiologically relevant loading, with and without simulated ligament contributions, in order to elucidate the effects of material pairings. A cruciate retaining TKR implant was created by combining a 3D printed ABS plastic tibial component with the standard cobalt chrome femoral component, as well as a 3D printed ABS plastic replica femoral component. This results in both M-P and P-P articulations that were mounted to a VIVO 6-DOF joint motion simulator (AMTI, Watertown, MA), which was used for in vitro constraint testing using functional laxity tests. Anterior-posterior (AP) and internal-external (IE) constraint was measured based on resulting deviations from the normal path when superimposed AP and IE loads were applied. Ligaments were simulated as tension-only point-to-point springs using the soft tissue modelling capabilities of the VIVO. Different kinematics were observed between the M-P and P-P implants which could be the result of different initial implant positioning on the joint motion simulator or due to “stiction” of the P-P implant. The functional laxity of the implant system tested appears to be relatively insensitive to the material pairing and ligament presence. These relationships are complex and hard to predict, which underscores the importance of pre-clinical in vitro testing.
Multisensory integration is driven by a process of sensory reweighting during which each input is assigned a weight depending on the current functional state of a particular sensory system, the task itself and the context in which it is being performed. The primary aim of this study was to determine which of the two inputs between ankle proprioception and vision is upweighed during a postural control task when the two inputs provide conflicting information pertaining to direction of body sway. Achilles tendon vibration and visual flow were used to create sensory conflict, which produced center of pressure (COP) sway in opposite directions when applied independently. The baseline conditions (1) consisted of eyes open quiet stance condition, eyes closed with vibration applied on the Achilles tendons (2) and eyes open with visual flow (3). The experimental condition simultaneously combined vibration and visual flow. COP excursions were recorded in 10 healthy young adults to evaluate the magnitude and direction of sway produced by vibration and/or visual flow. Additionally, lower body joint kinematics were evaluated to understand the multi-segmental strategies and their adaptation to the various sensory manipulations. The results showed that visual flow moderated the extent of backward COP and ankle angular displacement produced when vibration was applied independently. Additionally, visual flow was also found to reduce the extent of predominant hip strategy generated by ankle vibration. The findings show that visual input plays a significant role in maintaining stability and that ankle proprioception is downweighed during conflicts between vision and proprioception. This has important implication for balance training using controlled visual flow in patients with balance disorders and elderly.
Accelerometers have become extremely popular in the measurement of stride frequency as well as other related stride variables with current sensors capable of recording both accelerations and electromyography. The purpose of this preliminary investigation was to assess the estimation of stride frequency during running using a single tri-axial accelerometer compared to a commonly used infrared device the OptojumpTM system. Five healthy participants wore a Delsys Trigno tri-axial accelerometer attached to the right anterior shin and participants repeatedly ran at a submaximal pace through a four metre section of OptojumpTM. Stride frequency was calculated as stride time divided by one. For the OptojumpTM, stride time was the sum of contact and flight times from two consecutive steps. For the accelerometer, stride time was calculated as the time between two consecutive foot contacts on the right side. Foot contact was identified by local maxima in the Y (medial-lateral) acceleration trace. Estimates of stride frequency were compared using paired samples t- tests, intraclass correlation coefficients (ICCs) and Bland and Altman 95% limits of agreement (LOA) with significance set at p < 0.05. The mean difference between estimates was 0.01 Hz (95% LOA: -0.05-0.07 Hz) with single and average ICCs for stride frequency of 0.93 and 0.96 respectively. The results suggest that an accelerometer attached to the shin can accurately estimate stride frequency in running. Discrepancies in stride frequencies can be partially explained by differences in device sampling rates i.e. 137.15 Hz versus 1,000 Hz
The Force and Motion Foundation Updates...
The Force and Motion Foundation
The Force and Motion Foundation is proud to announce that our current $10,000 scholarship will be awarded in cooperation with the Orthopaedic Research Society (ORS).
Please visit http://www.ors.org/force-motionors-young-scientist-scholarship/ for more information on the Force & Motion Foundation/ ORS Young Scientist Scholarship.
2016 4th Quarter recipients of the travel awards are: Robin Healy, Ryo Ueno, Melissa Mansfield
Submit your 2017 1st Quarter Scientific Poster NOW for the F&M $500 Travel Scholarship!
Please check back in the future for information on more scholarship offers